Learn how to operate the Dissolution Test (DT) Apparatus.
Sample Interval Setting:
Parameter setting from OPTN key:
Wakeup Time Setting:
Circulating Pump Setting:
Procedure for USP apparatus I method (basket):
Procedure for USP apparatus II method (paddle):
Resetting the Instrument:
Halting / Aborting Test:
Power Fail Condition:
Sample Withdraw Procedure:
Dissolution Medium Preparation:
Share this article with your friends
Do you like this post? Link this article to your website/blog by copying the HTML code below.