To describe the procedure for operation and calibration of Sonicator.
This SOP is applicable for operation and calibration of Sonicator.
Officer/ Executive - Quality Control
Manager - Quality Control
5.1.1 Ensure that the area and instrument is clean.
5.1.2 Fill the purified water in both on the optimum level.
5.1.3 Switch “ON” the main power supply and press “POWER” button to “ON” .
5.1.4 Slowly keep the lower tray in to bath.
5.1.5 Select the SONIC by MODE button and control the strength of sonic.
5.1.6 Select “TIME” by “MODE” button and set the operating time by “UP/DOWN” button.
5.1.7 Select “TEMP” by “MODE” button and set the temperature by “UP/DOWN” button.
5.1.8 After feeding all parameter, push the “START/STOP” button to start cleaning.
5.1.9 When the items are removed from the sonicator, press “START/STOP” button and “POWER” button.
6.1 SOP - Standard Operating Procedure
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
.moc.enilediugamrahp@ofni :liamE Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩