To check operation of the equipment for reliable and accurate results.
This procedure is applicable to disintegration test apparatus, installed in quality control department.
3.1 Doing : Technical Assistant/Executive
3.2 Checking : Executive /Assistant Manager
Head of the Department
Frequency : Once in a month
5.1 Operate the instrument as per standard operating procedure.
5.2 When indicator display 37°C, measure the temperature of individual beaker with thermometer and record the results in Annexure- I
5.3 Also record the number of up down cycles per minute. Note down four reading for each bath and record the results in Annexure –I.
5.4 Also measure the oscillation and record the results in Annexure –I.
5.5 If the instrument is out of limit as in Annexure-I, follow the S.O.P.
Note : Each instrument is calibrated with the respective serial number bath. Do not interchange the bath.
6.1 °C = Degree centigrade
6.2 no. = Number
QUALITY CONTROL LABORATORY
OPERATIONAL CHECK REPORT OF DISINTEGRATION TEST APPARATUS
Temperature setting : 37°C Observed temp. Beaker 1. _________ ( Limit : 35.0°C to 39.0°C)
Beaker 2. _________ ( Limit : 35.0°C to 39.0°C)
Observed Up down cycles/minute
Limit : 29 to 32 cycles/minute
Measurement of Oscillation (Stroke length )
Limit : 55 +/- 2 mm
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
.moc.enilediugamrahp@ofni :liamE Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩