1.0 OBJECTIVETo clearly define the cleaning of Metrohm Karl fisher.
2.0 SCOPEThis procedure is applicable to karl fisher installed in quality control laboratory.
3.0 RESPONSIBILITY3.1 Doing : Technical Assistant
3.2 Checking : Executive /Manager
4.0 ACCOUNTABILITYHead of the Department
5.0 PROCEDURE5.1 Before switching ‘ON’ the instrument wipe the instrument with cloth.
5.2 On completion of the titration, Press ‘Out’ at the rear side of the Ti stand to drain out the solvent from
the titration vessel.
5.3 Press ‘In’ at the rear side of Ti stand to fill the vessel with Methanol.
5.4 Again drain off the methanol.
5.5 Switch off the instrument
5.6 After switch off the instrument, clean the instrument with cotton cloth.
Related: Calibration of Karl Fischer Apparatus
6.0 ABBREVIATION6.1 SOP - Standard Operating Procedure
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
Email: .moc.enilediugamrahp@ofni Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩