1.0 OBJECTIVETo check the performance of low temperature water bath for reliable and accurate results.
2.0 SCOPEThis procedure is applicable to Low temperature water bath installed in Q.C.Dept.
3.0 RESPONSIBILITY3.1 Doing : Technical Assistant/Executive
3.2 Checking : Executive /Manager
4.0 ACCOUNTABILITYHead of the Department
5.0 PROCEDUREFrequency :Daily
5.1 Operate the instrument as per respective standard operating procedure.
5.2 Fill D.M. water in water bath.
5.3 Switch ‘ON’ the mains.
5.4 Switch ‘ON’ the mains of instrument.
5.5 Set temperature 25°C using arrow key.
5.6 Observe the set temperature on display for 30 min. and record the results in Annexure-I
5.7 Repeat the procedure for three times.
5.8 Also note down the results with calibrated thermometer and record the results in Annexure-I.
5.9 If the performance of the instrument is satisfactory and set temperature obtained within limit, otherwise follow the S.O.P.
Related: SOP for Water Bath
6.0 ABBREVIATIONS6.1 °C = Degree centigrade
PERFORMANCE CHECK REPORT OF LOW TEMPERATURE WATER
Set temperature :________ Thermometer Id.No.__________
Acceptance criteria : + 0.2oC from set temperature
Temperature on Dispaly
Observed temperature with
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
Email: .moc.enilediugamrahp@ofni Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩