5.1 FOR MICROSCOPE
5.1.5 Place the specimen slide on the stage.
5.1.6 Rotate the nosepiece to move the 10 X objective into working position.
5.1.7 Lower the 10 X objective by rotating the ‘coarse adjustment knob’ to its positive stop.
5.1.8 Adjust the condenser in sharp focus using the “Condenser focusing knob”.
5.1.9 Sharpen the focus using “fine adjustment knob” .
5.1.10 Adjust proper field & high power 40 X objective or 100 X oil emulsion objective, by rotating the “coarse adjustment knob”.
5.2 PHOTOGRAPHY5.2.1 After adjusting proper field of specimen in sharp position, adjust the attachment of camera in computer.
5.2.2 Open the directory ATI multimedia.
5.2.4 Then click on the button marked “Auto scan”, “OK”.
5.2.5 Click the tuner mode, for screen.
5.2.6 Click VHS connector.
5.2.7 Click capture still for taking photograph after setting proper field by fine adjustment of microscope.
5.2.8 Click the print image, for getting printout.
5.2.9 After completion of work, close the ATI player and switch ‘OFF’ the microscope.
Gift this article to your friends - Share it !!!
Do you like this post? Link this article to your website/blog by copying the HTML code below.