Place the ultrasonic bath on a leveled surface.
Ensure that all the connections of the instrument are proper.
Ensure that water level is adequate.
Ensure that filled water is clear.
Switch ‘ON’ the mains.
Switch ‘ON’ the ‘ULTRASONIC’ bath button.
Set timer in clockwise direction.
If method of analysis requires simultaneous heating also, then only switch ‘ON’ the ‘HEATER’.
Never turn the timer in anticlockwise direction.
Note: Change the water in the ultrasonic bath once a week or early if required.
Operating instructions manual of ultrasonic cleaning bath.
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
.moc.enilediugamrahp@ofni :liamE Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩