Method APrepare the solution of the substance under examination (test solution) as directed in the monograph. Prepare not fewer than three standard solutions of the element to be determined, covering the concentration range recommended by the manufacturer of the instrument for the element to be determined and including the expected value in the test solution. Any reagent used in the preparation of the test solution should be added to the standard solutions in the same concentration. After calibrating the instrument as directed above, introduce each standard solution into the flame three times, and record the steady reading, washing the apparatus thoroughly with water after each introduction. Between each measurement a blank solution should be aspirated and the reading should be allowed to return to zero level. If a furnace is used, it is fired between readings.
Method BPlace in each of not fewer than three similar volumetric flasks equal volumes of the test solution as directed in the monograph. Add to all but one of these flasks a measured amount of the specified standard solutions containing steadily increasing amounts of the element being determined. Dilute the contents of each flask to the required volume with water.
Gift this article to your friends - Share it !!!
Do you like this post? Link this article to your website/blog by copying the HTML code below.