To describe the procedure for operation of Magnetic stirrer with hot plate.
This SOP is applicable for operation Magnetic stirrer with hot plate.
Officer/ Executive - Quality Control
Manager - Quality Control
5.1.1 Ensure that the instrument is clean and free from any dust particle.
5.1.2 Connect the main supply of the instrument.
5.1.3 Switch ‘on’ the instrument.
5.1.4 If required adjust the required temperature by temperature ‘knob’.
5.1.5 Keep the beaker (Having Sample) on the plate and wait for heating.
5.1.6 Deep the magnetic stirrer in beaker and start the stirring by knob.
5.1.7 Stirring speed should be control by ‘knob’ to slow, middle and high.
5.1.8 Switch ‘off’ the instrument after completion of work
5.2.1 Never keep the heater in ‘on’ condition, when not in use.
6.1 SOP - Standard Operating Procedure
6.2 No. - Number
6.3 QA - Quality Assurance
6.4 NA - Not Applicable
6.5 CCF - Change control format
6.6 QC - Quality Control
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
Email: .moc.enilediugamrahp@ofni Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩