1.0 OBJECTIVETo clearly lay down the procedure for cleaning
2.0 SCOPEThis procedure is applicable to friability test apparatus installed in Quality Control Department.
3.0 RESPONSIBILITY3.1 Doing : Technical Assistant
3.2 Checking : Executive /Manager
4.0 ACCOUNTABILITYHead of the department.
5.0 PROCEDURE5.1 After completion of friability test, switch “OFF” the instrument.
5.2 Switch “OFF” the main power supply.
5.3 Remove the knob on the shaft by pressing the button on the knob and pull the knob out.
5.4 Remove the lid of drum and clean the lid with tissue paper.
5.5 Clean the drum with tissue paper and make it visually clean.
5.6 Sonicate the sieves in sonicator for about 5 to 10 mins.
5.7 Remove the sieves from sonicator.
5.8 Dry with in air.
Related: Calibration of Friability Test Apparatus
6.0 ABBREVIATIONS6.1 SOP - Standard Operating Procedure
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
Email: .moc.enilediugamrahp@ofni Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩