To describe the procedure for calibration of Slit – To – Agar air sampler.
This SOP is applicable for calibration of Slit – To – Agar air sampler installed in microbiology department.
3.1 Doing : Technical Assistant(QC)/Executive
3.2 Checking : Electrical Engineer (
Head of the department.
Frequency : Once in a year.
5.1 There are three levels, necessary to calibrate the Slit – To – Agar air sampler.
5.2 LEVEL – 1
5.2.1 Check calibration with air velocity.
5.2.2 Take sample of air and monitor its air velocity with anemometer and record 8 reading and take average.
5.2.3 Calculate volume in CFM by multiplying outer area with average air velocity.
5.2.4 Record the result.
5.3 LEVEL – 2
5.3.1 This test calibration is accomplished by removing the battery voltage and replacing the battery with variable power supply, and observing at what voltage the low light illuminate.
5.3.2 The use of a fluke 73 series volt meter or equal is utilized to accomplish this task.
5.3.3 The low voltage warning light should turn on at approximately 10.5V + 3%. Record test data.
5.4.1 General inspection of portable Slit – To – Agar air sampler light, switches, and battery should be done at this time and it should be noted.
% = Percentage
CFM = Cubic feet per minuteLPM = Litre per minute
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
.moc.enilediugamrahp@ofni :liamE Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩