1.0 OBJECTIVETo describe the procedure for Operation and calibration of water bath.
2.0 SCOPEThis SOP is applicable for Operation and calibration of water bath.
3.0 RESPONSIBILITYOfficer/ Executive - Quality Control
4.0 ACCOUNTABILITYManager - Quality Control
5.1 Operation5.1.1 Ensure that the instrument is clean and calibration of temperature indicator is within due date of calibration.
5.1.2 Fill and check the water level, if required fill purified water to the acceptance level. The minimum water level is indicated by a black line on the water level indicator on left.
5.1.3 Switch “ON” the ring both by pressing “ON/OFF” switch.
5.1.4 The digital temperature controller cum indicator will indicate the actual temperature of water.
5.1.5 Set the desired temperature by pressing the PRESS to SET switch and adjusting the SET pot.
5.1.6 Approximately 1ºC before the set temperature, the heater will start going on and off Heater action is indicated by the LED on the DTC. Allow a few minute for the temperature to stabilize.
5.1.7 If the indicated temperature is less than desired, turn the fine tuning pot on the DTC clockwise or vice versa.
5.2 Precautions5.2.1 Do not operate without water.
5.2.2 Switch OFF when instrument is not in use
5.3.1 Calibration after three month.
5.3.2 Set the desired temperature against standard thermometer at 37 ºC, 50 ºC and 90 ºC.
Standard value (in ºC )
Set value (in ºC )
6.0 ABBREVIATIONS6.1 SOP - Standard Operating Procedure
6.2 ºC - Degree centigrade
6.3 DTC - Digital temperature controller
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
Email: .moc.enilediugamrahp@ofni Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩