1.0 OBJECTIVETo clearly lay down procedure for cleaning of fourier transform infrared spectrophotometer.
2.0 SCOPEThis procedure is applicable to clean the fourier transform spectrophotometer installed in quality control department.
3.0 RESPONSIBILITY3.1 Doing : Technical Assistant/Executive
3.2 Checking : Executive/Manager
4.0 ACCOUNTABILITYHead of the Department
5.0 PROCEDURE5.1 Clean the instrument with clean cloth.
5.2 For DRS system
5.2.1 After each I.R , discard the sample and wipe the sample cap with tissue paper.
5.3 For liquid samples
5.3.1 After each IR ,wash the KBr pellet with carbon tetrachloride.
5.4 Switch ‘OFF’ the instrument and wipe it with clean cotton cloth.
5.5 Record in the format of instrument log card.
Related: Calibration of FTIR
6.0 ABBREVIATIONS6.1 I.R = Infra red
6.2 S.O.P. = Standard operating procedure
6.3 KBr= Potassium Bromide
Ankur Choudhary is India's first professional pharmaceutical blogger, author and founder of Pharmaceutical Guidelines, a widely-read pharmaceutical blog since 2008. Sign-up for the free email updates for your daily dose of pharmaceutical tips.
.moc.enilediugamrahp@ofni :liamE Need Help: Ask Question
Spread the Knowledge ⇩⇩⇩